The regulation concerns continuous totalisers produced and imported to Norway.
Regulation on requirements to continuous totalisers
Continuous totalisers used for landing fish must have a display or other clear reading option placed in or by the weighing instrument.
The measured quantity must be able to read continuously. When the weighing instrument is in use, the weighted quantity must be able to read directly.
The weighing instrument must have at least one additional display that repeats the display of the weighing result from the main display. The additional display must not have functions that can affect the weighing result. The additional display must have the same division as the main display.
The weighing instrument must have more than one counter, one being a continuous counter that runs continuously and which cannot be reset without breaking seals. The total counter shall have the same division as the basic counter.